There are currently no product reviews.
The manual was complete and of great quality. Originally a tri-lingual file, I first received only one language. After a note to owners-manuals.com, I quickly received the remaining languages... Great service, definitely worth it.
Thank you for providing this manual and at low cost.
The Philips scope is of excellent quality, longevity and build and had a couple of faults
when it was passed to me. Having the CCT diagrams is a blessing.
I have fixed the problems and also modded the scope to my requirements.
I have built a 24v Li-Ion pack for portable use from old but good laptop batts.
it is working beautifully and I am well pleased. Keep up the good work Guys.
manual was very helpful in learning how to propery use my washer. I could not find this manual anywhere else.
Awesome quality manual. You really saved my bacon with this one. Was looking for some specific information with regards to my "new" vintage VCR that didn't come with the owners manual. This site is truely a goldmine of available manuals. The quality of the scans are top notch.
Thank-you so much for this awesome manual. If you're looking for this Sony SL-HF400 owners manual, this is the one you NEED to buy. Definitely worth the money.
The manual was made available promptly. I is a clean scan of the original. I had no problem downloading it. The scan was well centered and cleanly formatted. It is as good a product as can be had without being the original document.
Move DY so that the rear edge position of DY comes to the center of the two marked lines and fasten DY as d=dâ (Figure l-3-4). Further insert the rubber wedge between DY and CPT funnel from the top and raise DY backwards.
a-aâ, b-b Show the geometric center line of CPT Screen bâ FIGURE l-3-4 b
a - aâ b - bâ
Geometric line of CPT
center - a
C-F MAG P#2773672
The openings of the purity magnet should be symmetric: on the right and left sides (P#2773671) on the upper and lower sides (P#2773672) FIGURE l-3-2 (7) Receive the single red signal. When the single red signal is not available, short circuit between the base and emitter of Q845, and between the base and emitter of Q857 and set to single red. (8) Pull back DY gradually and when the color unevenness of both sides of the picture disappear, mark the rear edge position of DY on the tape wound around CPT neck as shown in Figure l-3-3. Pull back DY further and just before the color unevenness of both sides of the picture disappear, mark the rear edge position of DY on the tape in the same way. At this time, pull back DY so that the center axis of DY and CPT axis match.
(10) Set CPT axis direction magnetic field of the artificial magnetic field according to the artificial magnetic field setting list classified by destination. (The direction of the magnetic field should be from the CPT screen side to the neck side.) (11) After degaussing it from outside, check the purity in each color to R, G and B visually. Then, turn the screen to white and check the landing at the screen position shown in Fig. l-3-5 with a microscope. Criteria with microscope There should be no miss landing at positions 2, 4, 8 and 10. Green beam should be at center of the green phosphor at position C. (Refer to the miss landing criteria.) (12) Turn over the direction of CP?axis direction magnetic field of the next artificial magnetic field and check it by the same way as item (11). The positions of miss landing criteria with a microscope should be 2, 4, 8 and 10 (Figure l-3-5).
MARK I NC
Miss landing criteria The following conditions are defined as miss landing. Each color beams shines on the phosphor of the applied color and there are phosphor parts which are not luminous (shaded parts in
FIGURE l-3-3 44